FIS Sensor Gas Chromatograph (SGC) realizes quick gas separation and super gas sensitivity by using FIS original short column and FIS semiconductor gas sensor.
Since extremely high sensitive semiconductor gas sensor is used as a detector, ppb (parts per billion) level measurement can be realized.
Simply injecting sample gas starts measurement automatically. The measurement is completed in 4 or 8 minutes.
SGC uses ambient air as carrier gas so that high pressure gas cylinder is not necessary.
Note: Some models need the cylinder.
Our original technology including a short length column has realized a small size (B4 size) and light weight (5.5kg) unit with excellent portability that does not require a limited installation site.
Automatic continuous measurement is possible by setting total measurement duration time or total measurement times.
Simple configuration using ambient air as carrier gas and highly sensitive semiconductor gas sensor as a detector realizes super sensitive measurement.
SGC can be used to detect very small gas in various fields.
SGC is supplied with a standard software displaying the real-time measurement results on PC. This software has various features such as chromatography chart, gas concentration display, and continuous measurement (optional) display.
* Please supply your own PC.
SGC can be customized to meet your demand.
|Product name||Sensor Gas Chromatograph|
|Measurement principle||Gas chromatography using semiconductor gas sensor|
|Target gases (*1)||Hydrogen sulfide
|Measurement concentration||Hydrogen sulfide: 5ppb to 1ppm
Methyl mercaptan: 5ppb to 1ppm
Dimethyl sulfide: 5ppb to 1ppm
|Ammonia: 50ppb to 100ppm
Trimethylamine: 50ppb to 10ppm
|Hydrogen: 1 to 150ppm (*2)
Carbon monoxide: 1 to 150ppm
Methane: 1 to 150ppm
|Acetone: 100ppb to 20ppm
Ethanol: 100ppb to 100ppm
Acetaldehyde: 100ppb to 100ppm
Isoprene: 100ppb to 20ppm
|Toluene: 5ppb to 1ppm
Ethylbenzene: 5ppb to 1ppm
Xylene: 5ppb to 1ppm
Styrene: 5ppb to 1ppm
|Minimum display resolution||0.1ppb or 0.1ppm|
|Warm-up time||5 to 60 min (*3)|
|Measuring time||4 min||4 min (*4)||4 min||8 min||8 min|
|Sampling gas amount (*5)||2cc||2cc||2cc||5cc||5cc|
|Measurement results||On PC display (*6)|
|Sample injection||Manual injection with a syringe. Auto-start of the measurement (*7)|
|Carrier gas||Pumped and filtered clean ambient air (*8)||Cylinder air|
|Signal output via||RS232C (*9)|
|Power supply||100 to 240V AC|
|Power consumption||Approx. 40VA|
|Measurement||260(W) X 135(H) X 340(D) mm||260(W) X 135(H) X435(D) mm|
|Operating temp/humid||Temperature: 10 to 30C Humidity: 20 to 80%RH (No dew condensation)|
|Storage temp/humid||Temperature: -20 to 60C Humidity: 20 to 80%RH (No dew condensation)|